DIGITAL VOLTAGE PHASING METER
? Instrument for measuring overhead and underground voltages.
? Three modes of operation AC, Hi-Pot, and DC.
? 0-40KV meter up to 240KV with resistors.
? Automatically adjust voltage display to match resistors if used.
? Hi-Pot mode for testing URD Cable.
? In Hi-Pot mode bar graph marks the peak voltage while cable is being charged.
? 0-4KV DC.
? Large easy to read digital meter and analog bar graph.
? Meter reads to nearest 100 volts.
? Meter has lighted background that can be turned on for use in poor light conditions.
? Meter enclosure made of high quality rugged extruded aluminum.
? Auto shutdown after 40 minutes if no voltage is detected.
? Long battery life from three AA’s.
? Auto self test and battery check during start up.
? The coiled, self retracting cable is 16” retracted and 10’ when fully extended.
? Unit includes: Digital volt meter, conductor hooks, carry case, batteries, and instructions.
CAT NO DESCRIPTION LBS
6702 0-40KV DIGITAL VOLT METER 5.0
6703 PAIR OF EXTENSION RESISTORS FOR USE WITH MODEL 6702? ?1.3
6702-1 ONE 15/25KV LOADBREAK BUSHING ADAPTER 0.6
6702-3 ONE 35KV LOADBREAK BUSHING ADAPTER 0.5
6702-2 ONE D.C. HIPOT ADAPTER FOR USE UP TO 35KV 0.5
A30395 REPLACEMENT CARRYING CASE 2.1
6711 KIT INCLUDES 6702 WITH CASE, (2) 6’ UNIVERSAL FIBERGLASS POLES WITH CARRYING CASE? 10.0
EXTENDED VOLTAGE RANGES FOR PHASING METERS
May be used on higher voltages with additional pairs of extension resistors. These extension resistors must be used in pairs to increase the voltage range.
NUMBER OF RESISTOR PAIRS??RANGE
0? ? ? ?40KV
1? ? ? ? 80KV
2? ? ? ? 120KV
3? ? ? ? 160KV
4? ? ? ? 200KV
5? ? ? ? 240KV
D.C. HIPOT ADAPTER
? The D.C. HiPot adapter is an encapsulated high voltage rectifier for use on the phase tell meters 6702/6706. When used in conjuction with the phase tell it will enable the operator to perform the following tests on systems up to and including 35KV phase to phase:
1. Test sections of underground cable to determine the integrity of insulation, i.e., failed cables.
2. Determine if all grounds have been removed after completion of working on a de-energized underground system.
? The HiPot adapter housing is fiberglass, 13” in length. A bell shaped female end allows it to be coupled to the end of the phase tell after the conductor hook is removed. The appropriate URD bushing adapter can then be installed on the other end of the HiPot adapter .
CAT NO DESCRIPTION LBS
6702-2 D.C. HIPOT ADAPTER 0.5
INSTRUCTIONAL VIDEO AVAILABLE UPON REQUEST